Manufactured in Poland
Rapid SARS-CoV-2 Antigen test
If you need a quick and inexpensive coronavirus detection test that has a sensitivity of 90%, the SARS-CoV-2 antigen test is the one you are looking for!
Only for professional in vitro diagnostics.
1. Take a sample from throat or nasal cavity.
2. Insert the swab into the tube
2.1 Squeeze the flexible tube to extrude the specimen from the head of the swab.
2.2 Take out the swab
3. Reverse the specimen collection tube, extrude 3 drops of the prepared specimen into the specimen well (S), then wait 10-15 minutes.
4. Interpret the results of SARS-CoV-2 antigen test
*Instructions show how to perform the test for a throat sample. Detailed instructions can be viewed in the downloads section at the end of the page.
An explanatory video showing how to perform the SARS-CoV-2 antigen test
Interpretation of the results of SARS-CoV-2 antigen test
Positive (+): Two colored lines appear. One colored line should always appear in the control line region (C) and another line should be in the T line region.
*NOTE: The intensity of the color in the test line regions may vary depending on the concentration of SARS-CoV-2 present in the specimen. Therefore, any shade of color in the test line region should be considered positive and recorded as such.
Negative (-): One colored line appears in the control line region (C). No line appears in the T line region.
Internal procedural controls are included in the test. A colored line appearing in the control region (C) is an internal procedural control.It confirms sufficient specimen volume and correct procedural technique. Control standards are not supplied with this kit; however,it is recommended that positive and negative controls be tested as a good laboratory practice to confirm the test procedure and to verify proper test performance.
INVALID: Control line (C) is still no line during the whole testing procedures. Insufficient buffer volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the procedure with a new test device. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
- Foil pouches, with test cassettes and desiccants
- Assay buffer
- Sterile swab and specimen collection tube
- Instruction for use
Materials Required But Not Provided
- Specimen collection container
The SARS-CoV-2 Antigen Rapid Test is for detection of SARS-CoV-2 antigens. Anti-SARS-CoV-2 monoclonal antibodies are coated in the test line and conjugated with the colloidal gold. During testing, the specimen reacts with the anti-SARS-CoV-2 antibodies conjugate in the test strip. The mixture then migrates upward on the membrane chromatographically by capillary action and reacts with another Anti-SARS-CoV-2 monoclonal antibodies in the test region. The complex is captured and forming a colored line in the Test line region.
Note that a negative test result may occur if the level of extracted antigen in the sample is less than the sensitivity of the test or if a poor quality sample is obtained.